Room with an oxygen production device (concentrator) for the hospital. An oxygen concentrator is a device that concentrates and filters oxygen obtained from the environment, thanks to which it becomes an infinite source of this element in a form ready for administration to the patient. The oxygen supplied by this oxygen concentrator never ends (it is concentrated on a regular basis) and the device can work in a continuous mode, depending on hospital needs.
FAN AND MEDICAL GAS STORAGE
An additional container where the central ventilation and air conditioning system for the hospital is located.
It is also a room for storing medical and technical gases such as: oxygen, nitrous oxide, nitrogen, compressed air for the needs of the entire hospital facility.
WATER TREATMENT AND PURIFICATION STATION
The station consists of a 3-stage water purification system through the use of 3 filters and biological water treatment (disinfection) through a filter with a UV lamp.
Stage I – a filter that filters water from mechanical (sedimentation) impurities. The filter removes from the water small fractions of solid impurities (sludge, sludge, etc.) larger than 20 microns.
Stage II – active bed insert for iron removal of water. It effectively retains iron compounds dissolved in water. Contains special fine-grained granules with high iron ion capture capacity.
III Degree – a granulated carbon cartridge removes chlorine and its derivatives from water, further improves its taste and smell.
Biological water treatment, i.e. to get rid of all bacteria, microorganisms and microorganisms, is done using a UV lamp filter. The light generated by her freely penetrates the flowing water. Strong ultraviolet rays eliminate all microbes, viruses found in the liquid.
Soiled Disposal Storage
A room isolated from the rest of the hospital, used for emptying, disinfecting and storing ducks and swimming pools as well as storing dirty underwear. Equipped with a sink, washer-disinfector or device for decontamination.
ENERGY WAREHOUSE AND POWER GENERATOR
Connection of municipal power supply, but also container power generator, oxygen storage and production, LED lighting, low energy consumption, the possibility of using slurry tanks, drinking water tanks, photovoltaic and energy storage option.